Currently on the market of medical equipment, a large number of new players. Perhaps it was the result of policies pursued by the state of health. But Yesterday’s radio producers, harvesters and ships quickly rebuild its facilities to adequately learn the funds for this project. To put it bluntly: to organize the production of quality products have not at all, as evidenced by, medical furniture, which breaks down the day after delivery. Today we’ll talk about how this huge number of models and manufacturers of beds to choose that a functional bed, which is needed to your patients. For a start we divide all the beds in the bed of imported and domestic products.
With regard to imports of beds, then in this sector than the issue price as Typically, other issues arise. This is the bedding-known European manufacturers, although in fairness, it should be noted that Asian and beds are good quality. With respect to the beds, which produced in our country today, I want to talk separately. So what to look for when choosing a functional bed? The very first thing, of course, is the author (ie, producer). With the manufacturer of licensed rf Ministry of Health the right to production of functional beds, the conformity certificate for the group 94 52 00, the registration certificate is a guarantee that you will get safe, quality product. With regard to structural features and components that are used in functional beds, then you need to pay attention to the following points: 1. Wheels – best if the wheels of imported producer with a gray rubber (do not leave marks on floor), preferably with a ball bearing (balls or rollers inside).
At least two wheels with brakes. Better if the wheel ‘TENTE’-leader in the manufacture of wheels for medicine. Attention is drawn to mount the wheels to the chassis – power block should not be bolted to the thin-walled profiles, because tube wall deformed, and the connection loose. 2. Frame – should be made of metal profile coated with polymeric powder paint. If the bed bed has changed in height, the joints must be screwed, and on ball bearings, it would eliminate backlash and scratches when lifting and lowering the bed. Section bed can be closed with plastic, sheet metal or metal mesh. The best option – a perforated plastic. Sheet metal, of course firmly, but in the processing and moisture port (bad dries) mattress cover is destroyed and as a consequence there unpleasant odor. Electric-grid – is well ventilated and processed, but must use polyurethane foam mattress. 3. Drive – depending on material capabilities and technical capabilities premises may be – Electric (better ‘LINAK’ or ‘DEWERT’), hydraulic, screw (it is desirable that the screw had overload protection). Screw should be made of steel and the nut is made of bronze. Just for drives used lockable pneumatic springs or other mechanical drive – comb (rostomaty). 4. Side rails – preferably during the whole bed with a reliable fixation in extreme positions. 5. Mattress – must match the number of beds moving sections, made of polyurethane foam and have removable hygienic cover. If guided by the principles set forth above, then you have a real chance to choose for your medical institution quality functional beds.